Poor responses in injection / aspiration
- Detach the injection holder from the tubing, as well as detaching the injection holder cap and silicone rubber gasket.
- Insert the supplied cleaning rod from the rear until it protrudes from the tip of the holder to push out particles of silicone seals or broken glass capillaries, and then reconnect the tubing.
How to connect the injection holder to PTFE Tubing.
Copyright © NARISHIGE Group. All Rights Reserved. 1999-2020